Emergency Pull/Push Station
Call Placed LED. Requires Single-Gang Electrical Box.
Manuals & Diagrams
Architects & Engineer's Specs
The Emergency Push/Pull Stations shall be shall be Alpha Communications® AlphaLinQ™ 100 series model EPS155, or approved equal. Operation of emergency signals shall be accomplished by a short pull on the nylon cord or by pulling / pushing down on the Red lever marked 'EMERGENCY".
The Red "CALL PLACED" L.E.D. indicator shall flash to assure the patient that the call has been placed. Switching mechanism shall consist of a reliable DPDT switch.
The EPS155 is typically used as a toilet emergency pull station and is not designed to be used in wet or damp environments. For wet or damp (or shower) type applications use the model# EPS337 station instead.
All electronic components needed for these features shall be mounted on a printed circuit board and all station components shall be mounted an a rugged ABS plastic faceplate. The station(s) shall be installed in a single gang ring with square box. Plug-in connectors shall be provided for connections. The EPS155 shall be UL® 1069 Listed to the latest standards.
Contractor shall observe all local and national electrical and building codes.