The call cord shall permit simple call origination and shall be acceptable for use in Oxygen, Geriatric or general applications.
The momentary contact, (non-powered) pneumatically activated switch shall be built into the plug and shall be activated by a slight pressure to the pad. An 8' (2.44m) long vinyl tube shall connect the pad to the 8-pin DIN plug. A stainless steel bed clamp shall be permanently affixed to the call cord, but shall be adjustable along the length of the cord. An integral 8-pin 'DIN' type connector shall be provided. The SF401GP shall be provided with a flattened 3.5" diameter soft vinyl pad with gripping feet on the bottom, to prevent it from slipping off of the patient's bed. The SF401GP shall be able to be activated by a slight pressure from the patient's hand, arm, body or chin.
Call cord shall be UL Listed 1069 and Listed to CSA C22.2 No. 205; UL File Number E61802
The Alpha Communications® / TekTone SF401GP Call Cord is for use with the NC300 and NC300II nurse-call systems where an 8-pin 'DIN' type jack is used at the patient bed station(s). Typically they are used with the IR319B and/or IR320B patient stations.
The SF401GP Call Cord is specifically designed for use in Geriatric and Oxygen environments, but may also be used in general applications. To place a call, the patient need only apply slight pressure to the pad using any part of the body. The SF401GP has a mechanical switch that is activated by slight pressure from the hand, arm or body and can even be positioned under the patient's chin. The round, flattened soft vinyl pad has gripping feet to prevent it from slipping off of the patient's bed. An adjustable bed clip is supplied for the SF401GP and is attached to the call cord.
Momentary contact, pneumatically activated
8-pin DIN plug
|Electrical:||Rated current 1/3 amp, 24 VAC-VDC|
Cord: 8' long (2.44m)