(shown with all 4 sections lighted)
Addressable corridor dome/zone light(s) shall be Alpha Communications® / TekTone model LI386LED or approved equal. The corridor light shall be sectional in design. Lights shall include LEDs to provide WHITE, RED, AMBER/YELLOW, and GREEN annunciation through a translucent plastic lens. All lenses shall be mounted on a FOG-WHITE flame retardant (UL® 94HB) plastic base panel. The four (4) colored LEDs shall indicate patient call priority and staff presence. Data signals from the dome/zone light shall be supervised to the central equipment (CE). The unit shall provide visual and digital communications to and from the CE. The LI386LED shall be a standalone unit that connects directly to the CE.
LI386LED shall be U.L. 1069 listed, when used with the corresponding U.L. 1069 nurse-call system.
Contractor shall observe all local and national electrical and building codes.
Height: 4.80" (122mm)
|Mounting:||Fits standard one or two gang electrical box with one or two gang plaster ring.
Base: FOG WHITE flame retardant (UL® 94HB) plastic. Lens: Flame retardant translucent polycarbonate plastic