The visual single patient station(s) shall be Alpha Communications® SF001JC, or approved equal and shall include a jack for the required model SF301/8 (single) call-cords or model SF302 (double) call-cords (or other specialty type cords as required).
The station shall be mounted into a standard single gang electrical box (or ring). Pigtail wiring shall be provided for call-placed and cord-out connections. Faceplate shall be White plastic (with matching screws provided).
Contractor shall observe all local and national electrical and building codes.
The SF001JC has a conventional 1/4" 'phono' type jack and accepts a variety of plug-in type call cords and/or devices. The SF001JC features a White Plastic faceplate, and mounts on a single gang electric box or plaster ring. It also provides for a 'cord-out' indication, when used with the AlphaEntry™/QwikBUS™ type systems (or equivalent).
|Dimensions:|| Height: 4.50" (114mm)
Width: 2.75" (69mm)
Depth: 1.50" (38mm)
Projects: 0.44" (11mm) from mounting surface
|Construction:||White plastic face plate|
|Mounting:||Steel City #52171 box with Steel City #52C14 plaster ring, or Alpha Communications® Model IH358 box with IH357 ring, or EXACT EQUAL. Vertical mount.|
|Connections:||Pigtail wiring (2 for call cord and 2 for cord-out)|