Corridor dome lamp shall be Alpha Communications® model LI406 or approved equal. The corridor lamp shall be sectional in design. Lamps shall include a WHITE, RED, YELLOW, BLUE, AMBER and GREEN flame retardant polycarbonate translucent plastic lens. All lenses shall be mounted on a FOG WHITE flame retardant polystyrene translucent plastic base panel. The lenses shall be individually removable without the use of tools for ease of bulb replacement. The six (6) bulbs in conjunction with the different colored lenses shall provide for indication of room status, when used with the AlphaStatus™ room status system(s), and associated RSS406 and/or RSS407 room status station(s).
Contractor shall observe all local and national electrical and building codes.
The Alpha Communications® LI406 sectional corridor lamp provides visual indication of calls originating from patient, staff, duty, bath, emergency, presence, and code call stations. Room status is indicated by RED, GREEN, YELLOW, BLUE, AMBER and WHITE colored sections. When used with the AlphaStatus™ room status system(s), and associated RSS406 and/or RSS407 room status station(s), the LI406 will indicate all calls originated from that room.
| Height: 6.56" (167mm)
Width: 4.56" (115mm)
Depth: 3.00" (75mm) from finished wall surface
||#1829 28v. 70mA (model LI028A)
||24VDC, 51mA (per bulb)|
Base: FOG WHITE flame retardant ABS plastic
|Mounting:||Requires special back box - contact us for details.|