The SF401P provides for call placement to the system station master(s) and for TV controls. Simple operation is provided by momentary switches.
To call a nurse simply press the pushbutton indicated " ". Separate controls are provided for TV ON/OFF and channel selection. TV sound is automatically silenced while the intercom is in use (at the patient station).
Pillow Speaker(s) shall be Alpha Communications® / TekTone model SF401P or approved equal. Pillow speakers shall have a 2.25" Mylar cone speaker. The pillow speaker case shall be constructed of ABS Flame retardant plastic. All momentary contact SPST 10 AMP switches shall be rated for over 1,000,000 operations. The pillow speaker cord shall be an 8 conductor cable and shall have an 8-pin 'DIN' plug. The cord shall also be provided with stain relief. Each pillow speaker shall be provided with a wall hanger, a security/bed clip and its own individual plastic bag. Controls shall be provided for entertainment, audio, intercom, nurse calling and channel selection of television, and shall be marked , TV respectively. The pillow speaker shall be Electro-Static-Discharge protected to over 20,000 volts.
The pillow speaker(s) shall be UL 1069 listed to the latest 4th Edition standards.