Master station function buttons are provided to select the system functions as follows:
The master station is also equipped with volume control for adjustment of incoming voice communication; tone switch for controlling the lowest priority tone signal only; beep switch for controlling the short tone signal heard when the keypad is in use; call light with indication of low, medium and high priority calls; CPE fault LED; and alphanumeric keypad used for entry of station numbers, zone and station selection, and for selection of menu functions.
The master station(s) also provide a parallel port and serial port for connection to UL1069® Listed devices. The system programming is accessible to the user via the keys on the master station or via the keyboard port which will accept any standard PC keyboard (similar to our model KB-1).
For the NC304LCD, incoming calls shall be displayed by room numbers in plain English on an integral LCD display with a minimum of 2 lines capable of displaying 40 characters each. The calls will display in order of time of origination, and in accordance with established priority. The master station(s) shall also provide a color VGA output to connect to a standard VGA monitor (our model# NC313) for color VGA call display. A parallel port and serial port shall be available for connection to UL1069® Listed devices. The user shall be able to access system programming via the keys on the master station(s) or via the keyboard port, which will accept any standard PC keyboard (similar to our model KB-1).
The master station(s) shall be equipped with an integral handset, volume control, color-keyed function switches (with and without LED's), audible signal control switches, call LED, busy LED, 2.50" speaker, electrostatic microphone, C.P.E. fault LED; and 12-button digital keypad for entry of station numbers, zone and station assignments, and for selection of menu functions. All buttons shall also switch to Alpha-character operation by depressing the switch marked "ALPHA", Alpha-character operation shall allow for alphanumeric room numbering, room labeling, and calling of rooms so designated. All switches shall be of spill-proof membrane switch design.
Each master station shall be able to address up to 512 single or dual stations per system, and up to 16 individual zones. The NC300II System shall support up to 8 master stations per system. The master station(s) shall be UL1069® listed.
Station housing shall be high-impact, flame retardant, ABS type plastic. Station(s) shall be desk mounted.
NC300II Systems shall meet all Veterans Administration specifications when used with additional Alpha Communications® equipment to be specified by Alpha Communications®.
|Dimensions (Console):||10.75" x 8.75" x 3.25" (274mm x 223mm x 83mm)
|Dimensions (Speaker):||2.50" (64 mm), 8 ohms
||Can address up to 512 single or dual patient stations, in up to 16 zones.
||0-80% relative humidity
||Plug-on connectors System|
|Protection:||All outputs are protected by selfresetting current-limiting devices|