Staff station(s) shall be Alpha Communications® / TekTone IR250, or approved equal. Model IR250 shall include solid state circuitry, combination yellow "CALL-PLACED" indicator and "CALL" button, combination red "IN-USE" indicator and "RESET" button. The staff station(s) shall provide for connection of smoke detectors, bath stations, and LI382 two bulb dome lights. The station shall provide signaling (via central equipment) to the appropriate zone lamp(s) using the specific flash rate on the dome light to indicate priority levels.
The station(s) shall utilize a 2.50" x 2.50" speaker/ microphone for accurate voice reproduction.
All station components shall be mounted on an attractive flame retardant type ABS plastic faceplate. Connections shall be plug-in be plug-in type.
||Height: 4.50" (119 mm) Width: 6.38" (162 mm) Depth: 2.75" (70 mm) Projects: 0.50" (12.7mm)
||Flame retardant type ABS plastic faceplate
||Light texture, fog white
||2.50" x 2.50", 60 ohm
||Alpha Communications® IH353 Box with IH352 Ring or Steel City #H3BD and #3GC plaster ring, or EXACT EQUAL. Horizontal mount.|