The interface for the Hill-Rom® SideCom® bed controls to Alpha Communications® / TekTone Nurse Call Systems shall be Alpha Communications® / TekTone RY351B or approved equal. The relay shall provide for direct interface to specific Alpha Communications® patient stations.
Wiring between specific Alpha Communications® patient stations and the RY351B shall be mounted on a single gang ring size sub-plate. The RY351B shall provide a non-resettable "ROUTINE" call at the nurses master station if the bed is unplugged.
The RY351B SideCom® adapter is designed for use with the IR311C, IR312C, IR319B and/or IR320B patient stations, and a HILL-ROM® SideCom unit. When used with the above patient stations and the HILL-ROM® SideCom® unit the RY351B adapter will provide all the necessary circuitry for a HILLROM® SideCom® unit to control all functions normally controlled by a pillow speaker. This includes but is not limited to nurse call and patient communications. Note: There is one RY351B SideCom® adapter required per SideCom® unit.
||Height: 1.25" (31.8mm) Width: 2.00" (50.8mm) Length: 4.25" (108mm)
||Circuit board mounted on a rugged aluminum mounting bracket.
||Reliable ramp lock plug-ins.|