Psychiatric stations shall be Alpha Communications® / TekTone Model IR316 or approved equal. Model IR316 shall include solid state LSI/microprocessor technology, yellow call-placed indicator, key switch, and combination red in-use indicator and reset button.
Activation of the key switch shall place a non-resettable call at the nurse master station and shall enable the staff emergency button on the associated staff emergency station. The key switch shall not deactivate the speaker / microphone station.
All station components shall be mounted on a rugged extruded aluminum panel. Connections shall be plugin type.
The IR316 shall be UL 1069 listed to the latest 4th Edition Standards. Contractor shall observe all local and national electrical and building codes.
Activation of the station key switch places a non-resettable call to the nurse master station and enables the emergency button on the staff emergency station (model SF336 or SF338) located in the ward. Call placement is indicated by the call-placed light on the station panel and at the associated dome lamp. The in-use light on the panel indicates that the ward is being monitored at the nurse master station via the speaker/microphone located in the ward. If a staff member should require emergency assistance, the staff emergency station (model SF336 or SF338) can be used to place a high priority call to the master.
When exiting the ward, the staff member will turn the key switch to the "OFF" position and push the reset button to deactivate the emergency button and cancel the call at the master station. The speaker/microphone will remain active should a staff member wish to monitor that particular area.
||Height: 5.00" (127mm) Width: 6.88" (175mm) Depth: 2.75" (70mm) Projects: 0.25" (6.5mm) above mounting surface
||Rugged, extruded anodized aluminum faceplate
||Brushed aluminum (clear anodized)
||Alpha Communications® model IH353 box with IH352 ring (or Steel City #H3BD with Steel City #3GC plaster ring), or EXACT EQUAL. Horizontal mount.
||Multiplex (using WM300B or WM300BP cable)|