IR301C Single Bed Station
The IR301C Single Bed Patient Station provides reliable two-way, hands-free communication between patient and nurse, and provides for programmable privacy, paging, and patient priority from the master station.
Calling the nurse is accomplished by pressing the call button on the end of the call cord. Solid-state LED's on the faceplate indicate the call has been placed and when the intercom is in use.
If the cord is accidentally removed from its receptacle, the nurse master station video monitor registers a noncancelable call with 'CORD OUT' indication. The call may be reset only by returning the call cord (or dummy plug) to its receptacle.
||Height: 4.50" (119mm) Width: 6.38" (162mm) Depth: 2.75" (70mm) Projects: 0.50" (12.7mm) above mounting surface
||Attractive, flame retardant ABS plastic
||Light texture, fog white color.
||Steel City #H3BD with Steel City #3GC plaster ring, or Alpha Communications® Part #IH353 box with IH352 ring, or EXACT EQUAL. Horizontal mount.
||Multiplex (using #WM300B or WM300BP cable)|
||2.50" x 2.50"|
Patient Station(s) shall be Alpha Communications® / TekTone IR301C, IR302C, or approved equal. Stations, and wiring from the Central Processing Equipment (CPE) to stations, shall be fully supervised. The patient station(s) shall include solid state. LSI/microprocessor technology, yellow call-placed indicator(s), and combination red in-use indicators and reset buttons. Single patient station(s) shall include one 0.25" jack receptacle for patient's call cord. Dual patient station(s) shall include two (2) call cord jacks.
The station(s) shall be use a 2.50" x 2.50" speaker/microphone for accurate voice reproduction. Removal of the call cord from the receptacle shall result in placing a call with the indication of 'CORD OUT' at the master station video monitor. The call can be reset only by replacing the call cord (or equivalent dummy plug) in its respectable.
All stations components shall be mounted on an attractive flame retardant ABS plastic faceplate, in fog white color. Connections shall be plug-in type.