Corridor dome lamp shall be Alpha Communications® / TekTone model LI380 or approved equal. The corridor lamp shall be sectional in design. Lamps shall include a WHITE, RED, YELLOW, and GREEN flame retardant polycarbonate translucent plastic lens. All lenses shall be mounted on a FOG-WHITE flame retardant polystyrene translucent plastic base panel. The lenses shall be individually removable without the use of tools for ease of bulb replacement. The four (4) bulbs in conjunction with the different colored lenses shall provide for indication of patient call priority and staff presence.
LI380 shall be U.L. 1069 listed to the latest 4th Edition standards, when used with the corresponding U.L. 1069 nurse-call system.
Contractor shall observe all local and national electrical and building codes.
The Alpha Communications® / TekTone LI380 sectional corridor lamp provides visual indication of calls originating from patient, staff, duty, bath, emergency, presence, and code call stations. Patient priority and staff presence are indicated by RED, GREEN, YELLOW, and WHITE colored sections. When used with any room station or IR310E Staff Station, the LI380 will indicate all calls originated from that room. When used with the IR300E Multi-Purpose Station or the IR315F Duty Station, the LI380 may be used as a zone light to indicate all calls from the zones to which those stations are assigned.
||Height: 4.56" (115mm) (overall) Width: 4.56" (115mm) Depth: 3.00" (75mm) from finished wall surface
||#1829 28v. 70mA (model LI028A)
||24VDC, 51mA (per bulb)
||Base: FOG WHITE flame retardant ABS plastic Lenses: Flame retardant translucent polycarbonate plastic (YELLOW, RED, GREEN and WHITE)
||Steel City #H2BD box with Steel City #2GC plaster ring, or Alpha Communications® Model #IH356 box with IH355 ring, or EXACT EQUAL.|