The SF301P provides for call placement to the system station master(s) and for TV controls. Simple operation is provided by momentary switches. To call a nurse or select television channels simply press the push button indicated " " or "TV" as desired. Volume control for TV only may be adjusted as necessary.
Pillow Speaker shall be Alpha Communications® / TekTone model SF301P (or SF301PE) or approved equal. It shall have a 2.75" Mylar cone speaker. The pillow speaker case shall be constructed of ABS Flame retardant plastic. All momentary contact SPST 10 AMP switches shall be rated for over 1,000,000 operations. The pillow speaker cord shall be standard telephone type 8 conductor cable and shall have a modular plug. The cord shall also be provided with stain relief. Each pillow speaker shall be provided with a wall hanger, a security/bed clip and its own individual plastic bag.
Controls shall be provided for entertainment, audio, nurse calling and channel selection of television, and shall be marked VOLUME, , and TV respectively.
The pillow speaker shall be Electro-Static-Discharge protected to over 20,000 volts.
The SF301P shall be UL 1069 listed to the latest 4th Edition standards.